MOS image sensors can generally handle a higher electrical charge compared to CCD image sensors, and are used under high light level conditions. CCDs feature low noise and are more suited for detecting low level light. MOS image sensors are also capable of operating on a 5 V supply which makes them easy to handle.
Prepare the following components.
1. Prepare the following components.Multichannel detector head or driver circuit (These products for some CCDs might not be available from us. See the image sensor datasheets or selection guide for detailed information.)
2. Multichannel detector head controller C7557-01 (when using a multichannel detector head): Allows easy data acquisition via USB connection to a PC
3. External power supply, A/D converter (Whether these are required depends on the multichannel detector head and driver circuit. See the datasheets.)
4. Frame grabber board (when using a multichannel detector head with a Camera Link interface)
Photosensitivity listed in the datasheets are typical values for product specifications and are not guaranteed values. Front-illuminated CCDs for example have somewhat large variations in spectral response characteristics from product to product because they have a structure where CCD gate electrodes are formed on a surface that receives incident light. Near the visible region, back-thinned CCDs have fewer variations in sensitivity compared to the front-illuminated type. In the ultraviolet region however, they exhibit large variations.
If spectral response characteristics for individual products are needed, then consult our sales office nearest you. Please note that spectral response characteristics depend on temperature. See our technical guide for information on the temperature dependence of spectral response characteristics.
We recommend first turning on the DC power to the CCD and then applying the power to the clock system. Turning on power in the reverse sequence or at the same time is not likely to harm the CCD.
It is recommended that the clock pulses overlap each other at 50%±10% of their amplitudes. Operation deviating from this condition may affect the saturation charge and charge transfer efficiency, so take this into account if high detection accuracy is required.
CCD characteristics vary with operating conditions. Operating conditions specified in the datasheet show the adjustment range. They must be adjusted within the specified range. We recommend using the CCD under typical operating conditions. Electrical and optical characteristics listed in the datasheet are the values when used under typical operating conditions.
When X-rays are directly incident on a CCD, the front-illuminated type is sensitive to X-rays from 0.5 keV to 10 keV, and the back-thinned type is sensitive from 0.1 keV to 10 keV. Front-illuminated CCDs coupled to an FOS (FOP coated with scintillator) or phosphor sheet are sensitive to X-rays of several ten keV or higher. See our technical information for details. If detecting X-rays at other energy levels, consult our sales office nearest you.
Datasheets contain a graph showing spectral response characteristics [vertical axis: quantum efficiency (QE)] and CCD node sensitivity (Sv). The voltage output at particular wavelengths can be calculated from the quantum efficiency and CCD node sensitivity. This calculation formula is as follows:
V = Number of incident photons × QE × Sv
When capturing two-dimensional images with an FFT-CCD, its operating principle requires a shutter or similar mechanism to block light so that no light strikes the CCD during charge transfer. (This is to prevent light striking the CCD during charge
transfer from being read out as a signal because the photosensitive area of an FFT-CCD also serves as a charge transfer register.) The signal produced by light incident on the CCD during charge transfer appears like vertical stripes in the acquired image. (This phenomenon is called smear.) If the charge integration time is extremely long compared to the readout time, then there is relatively little shading effect, so the desired image can be acquired even without a shutter.
When operating an FFT-CCD in binning mode, the light incident on the CCD during charge transfer is added to the signal by binning and is read out as one-dimensional data, so there is no problem if used without a shutter.
The inspection sheet conditions are shown on the datasheet. Compare the device using conditions close to those inspection conditions.
Abnormal phenomena that did not occur at the time of shipping inspection might occur due to damage (electrostatic discharge damage or deterioration caused by surge) during handling by the customer or due to unavoidable problems (caused by effects from cosmic rays, etc.) during shipping, so please consult our sales office.
There is no problem at all when using a front-illuminated CCD with no window under a vacuum of several pascals. However, at vacuum levels higher than approximately 10-2 pascals, the system may sometimes become contaminated from resin and so on, so please consult us. Back-thinned CCDs have also been used in vacuum, so please consult us if using them in vacuum.
Nearly all front-illuminated CCDs can be coupled to an FOP.
However, back-thinned CCDs cannot be coupled to an FOP. Please consult our sales office for more details.
To find the temperature range, see the operating temperatures and storage temperatures listed in the datasheet. The temperature and humidity must meet the conditions that will not cause moisture condensation on the sensor. Using at a humidity from 45% to 55% is recommended. Use in this recommended humidity range is also necessary as an ESD countermeasure.
Static electricity can destroy a CCD, so care is required during handling. We recommend using a workbench fitted with ESD
countermeasures, wrist straps, conductive mats, etc. Use of an ionizer is also recommended to remove static electricity that may be generated during handling. See the "Precautions" of CCD image sensors for detailed information.
See the "Precautions" of CCD image sensors for the instructions on how to store a CCD. After unpacking, we recommend storing in a low humidity desiccator.
Visible light cannot damage the CCD. However, ultraviolet light, X-rays and other high-energy radiation can cause damage.
See our technical information for details.
CCDs generate dark current even during readout operation. So even if the charge integration time is zero, there is still an output including dark current.
The packages mainly use ceramic materials, so manufacturing the package within a small tolerance is difficult. In view of quality control, the package tolerance is set to a large figure. Special processing techniques may make the package tolerance smaller, so consult us if needed.
Multichannel detector heads are not available for some CCD types, but simple evaluation circuit boards may be available for them. Contact our sales office for more information.
Custom specifications may be available at extra cost for some CCDs. Please contact our sales office if needed.
Types with no flaw or blemish may be available at extra cost for some CCDs. Please contact our sales office if needed.
Some CCD types with flaws or blemishes may be available. Please contact our sales office if necessary.
If that data can be prepared, then we will provide it for a fee.
It looks like you're in the . If this is not your location, please select the correct region or country below.
You're headed to Hamamatsu Photonics website for JP (English). If you want to view an other country's site, the optimized information will be provided by selecting options below.
For modern websites to work according to visitor’s expectations, they need to collect certain basic information about visitors. To do this, a site will create small text files which are placed on visitor’s devices (computer or mobile) - these files are known as cookies when you access a website. Cookies are used in order to make websites function and work efficiently. Cookies are uniquely assigned to each visitor and can only be read by a web server in the domain that issued the cookie to the visitor. Cookies cannot be used to run programs or deliver viruses to a visitor’s device.
Cookies do various jobs which make the visitor’s experience of the internet much smoother and more interactive. For instance, cookies are used to remember the visitor’s preferences on sites they visit often, to remember language preference and to help navigate between pages more efficiently. Much, though not all, of the data collected is anonymous, though some of it is designed to detect browsing patterns and approximate geographical location to improve the visitor experience.
Certain type of cookies may require the data subject’s consent before storing them on the computer.
This website uses two types of cookies:
There are two ways to manage cookie preferences.
If you wish to restrict or block web browser cookies which are set on your device then you can do this through your browser settings; the Help function within your browser should tell you how. Alternatively, you may wish to visit www.aboutcookies.org, which contains comprehensive information on how to do this on a wide variety of desktop browsers.
Occasionally, we may use internet tags (also known as action tags, single-pixel GIFs, clear GIFs, invisible GIFs and 1-by-1 GIFs) at this site and may deploy these tags/cookies through a third-party advertising partner or a web analytical service partner which may be located and store the respective information (including your IP-address) in a foreign country. These tags/cookies are placed on both online advertisements that bring users to this site and on different pages of this site. We use this technology to measure the visitors' responses to our sites and the effectiveness of our advertising campaigns (including how many times a page is opened and which information is consulted) as well as to evaluate your use of this website. The third-party partner or the web analytical service partner may be able to collect data about visitors to our and other sites because of these internet tags/cookies, may compose reports regarding the website’s activity for us and may provide further services which are related to the use of the website and the internet. They may provide such information to other parties if there is a legal requirement that they do so, or if they hire the other parties to process information on their behalf.
If you would like more information about web tags and cookies associated with on-line advertising or to opt-out of third-party collection of this information, please visit the Network Advertising Initiative website http://www.networkadvertising.org.
We use third-party cookies (such as Google Analytics) to track visitors on our website, to get reports about how visitors use the website and to inform, optimize and serve ads based on someone's past visits to our website.
You may opt-out of Google Analytics cookies by the websites provided by Google:
We inform you that in such case you will not be able to wholly use all functions of our website.