Currently, it is not possible.
Yes, as long as the microscope has a side port that allows 100% optical path switching. In that case, a C-mount adapter is required.
If the height of the center of the optical axis of the side port is between 65 mm and 105 mm, you can adjust the height of the MAICO® unit only.
You can use our "HCImage series", "HCImage Live" to control MAICO® only, and "HCImage Acquisition / Analysis" to control the focus of the microscope from the software.
"HCImage Acquisition/Analysis" can control the following controllers for Z drives.
・Ludl：MAC5000 / MAC6000
・Prior：OptiScan Ⅰ、Ⅱ、Ⅲ / ProScan Ⅱ、Ⅲ / ES10ZE
・Physik Instrumente (PI)：piezo E-709 / E-665 / E-662
For other details, please contact us.
It is available online, please visit: (https://dcam-api.com/dcam-sdk-login/)
You can choose any combination of the four wavelengths, but one of 405 nm or 488 nm must be installed. You cannot install more than two lasers of the same wavelength.
Currently no, however, we may increase should there be a demand for other wavelengths.
When comparing the standard type and the high-sensitivity GaAsP type in the 488 nm subunit, about 1.9 times more photoelectrons can be obtained. (EGFP, photosensitive from 510 to 540 nm)
Approximately 2.3 times more photoelectrons can be obtained when comparing the standard type with the high-sensitivity GaAsP type in the 561 nm subunit. (DsRed, 580 to 619 nm photosensitive)
When comparing the standard type with the high-sensitivity GaAsP type in the 638 nm subunit, approximately 2.3 times as many photoelectrons are obtained. (Alexa 647, 660 to 730 nm photosensitive)
No. The detector itself cannot be replaced, it would be necessary to replace the subunit.
The pinhole size can be adjusted in three steps for each subunit: Small, Medium and Large.
The smaller the pinhole size, the higher the resolution but the lower the amount of light detected. On the other hand, when the pinhole size is large, the resolution decreases while the amount of light detected increases.
Due to the balance between the amount of light detected and the resolution, it is recommended to set the pinhole size to Small when the objective lens is 20× to less than 40x, Medium when the objective lens is 40× to 63×, and Large when the objective lens is 64× or more.
Currently only fluorescence imaging is supported.
Yes, it is possible. The software allows you to select either simultaneous multi-wavelength measurement or sequential multi-wavelength measurement.
Zoom is a function that changes the magnification of the output image by changing the scanning range of the optical scanning element.
In general, zooming is expected to improve the resolution.
The actual resolution and magnification of the image depends on the objective lens you use.
It is 6.25 μm when the zoom function is 1×, and 3.125 μm when it is 2× (converted to 1× objective lens).
No, it is not suitable for FRET imaging. This is due to the optical path design that realizes simultaneous multi-wavelength observation without crosstalk.
According to international standards, devices equipped with lasers are classified into classes according to their light intensity and wavelength.
The classes range from 1 to 4, and subclasses are added to the alphabet. In general, the higher the class number, the more powerful the laser and the more extensive the protective measures. The largest class is 4, the next weakest is 3B, and the next weakest is 3R.
MAICO® is classified as 3R. Other common confocal laser microscopes are class 3B.
No, it is not necessary.
Yes, it is possible to obtain transmitted light images using a detector for fluorescence measurement.
The light intensity depends on the transmitted light source.
It is recommended to use MAICO® in "transmitted light mode" for transmitted light observation.
MAICO®'s superior features, including its subunit structure, are introduced here.
MAICO® enables imaging with reduced bleed-through between wavelengths, which is an issue in multi-wavelength simultaneous observation. We will introduce how we have achieved a reduction of bleed-through.
Explanation of the principle of a confocal microscope, which enables you to acquire an image that is less blurry, higher contrast, and higher resolution.
It looks like you're in the . If this is not your location, please select the correct region or country below.
You're headed to Hamamatsu Photonics website for GB (English). If you want to view an other country's site, the optimized information will be provided by selecting options below.
For modern websites to work according to visitor’s expectations, they need to collect certain basic information about visitors. To do this, a site will create small text files which are placed on visitor’s devices (computer or mobile) - these files are known as cookies when you access a website. Cookies are used in order to make websites function and work efficiently. Cookies are uniquely assigned to each visitor and can only be read by a web server in the domain that issued the cookie to the visitor. Cookies cannot be used to run programs or deliver viruses to a visitor’s device.
Cookies do various jobs which make the visitor’s experience of the internet much smoother and more interactive. For instance, cookies are used to remember the visitor’s preferences on sites they visit often, to remember language preference and to help navigate between pages more efficiently. Much, though not all, of the data collected is anonymous, though some of it is designed to detect browsing patterns and approximate geographical location to improve the visitor experience.
Certain type of cookies may require the data subject’s consent before storing them on the computer.
This website uses two types of cookies:
There are two ways to manage cookie preferences.
If you wish to restrict or block web browser cookies which are set on your device then you can do this through your browser settings; the Help function within your browser should tell you how. Alternatively, you may wish to visit www.aboutcookies.org, which contains comprehensive information on how to do this on a wide variety of desktop browsers.
Occasionally, we may use internet tags (also known as action tags, single-pixel GIFs, clear GIFs, invisible GIFs and 1-by-1 GIFs) at this site and may deploy these tags/cookies through a third-party advertising partner or a web analytical service partner which may be located and store the respective information (including your IP-address) in a foreign country. These tags/cookies are placed on both online advertisements that bring users to this site and on different pages of this site. We use this technology to measure the visitors' responses to our sites and the effectiveness of our advertising campaigns (including how many times a page is opened and which information is consulted) as well as to evaluate your use of this website. The third-party partner or the web analytical service partner may be able to collect data about visitors to our and other sites because of these internet tags/cookies, may compose reports regarding the website’s activity for us and may provide further services which are related to the use of the website and the internet. They may provide such information to other parties if there is a legal requirement that they do so, or if they hire the other parties to process information on their behalf.
If you would like more information about web tags and cookies associated with on-line advertising or to opt-out of third-party collection of this information, please visit the Network Advertising Initiative website http://www.networkadvertising.org.
We use third-party cookies (such as Google Analytics) to track visitors on our website, to get reports about how visitors use the website and to inform, optimize and serve ads based on someone's past visits to our website.
You may opt-out of Google Analytics cookies by the websites provided by Google:
We inform you that in such case you will not be able to wholly use all functions of our website.