X-ray image intensifiers have excellent sensitivity, but they have image distortion that should be corrected in image measurement. For CCDs, it is very difficult to manufacture devices with a large effective area. This is a disadvantage for X-ray imaging of large areas because, unlike visible light, optical lenses cannot be used to focus X-rays. On the other hand, a-Si devices can be manufactured in large sizes but have extremely long decay time, which may cause the problem of image lag during high-speed operation. Flat panel sensors were developed to solve all of these problems. Incorporating CMOS technology allows fabricating devices about mid-way in size between a-Si and CCD devices and that do not have image lag. The comparison table below shows the specifications of these devices.
|X-ray imaging device||Sensitivity||Resolution||Image size||Dynamic range||Cost performance||Distortion||Remarks|
|X-ray image intensifier||High||Low||Large||Very narrow||Good||High||Large camera size|
|CCD||Good||High||Small||Narrow||Low||Low||Difficult to produce devices with large area|
|a-Si||Low||Low||Very large||-||Low||Low||Long decay time & high price|
|Flat panel sensor||Good||Good||Large||Wide||High||Low||-|
Our standard process achieves a high aperture ratio of 79% in 50 × 50 µm pixel sizes and 87% in 100 × 100 µm pixel sizes.
This occurs because the flat panel sensor is accumulating an electrical charge before the software supplies an external trigger. So, exclude the first frame during measurement.
Although depending on the types of flat panel sensors, prepare a frame grabber board or network interface card, a signal cable, a power supply, and a PC, etc. For more details, see the datasheets.
We recommend using a monotonically increasing type series power supply equipped with a power transformer. If a switching power supply that generates high ripple noise is used, horizontal line noise may appear on the acquired images. Switching power supplies may also degrade the image quality when used in combination with a DC/DC converter and a LDO. Use a series power supply with low ripple noise which is within the allowable range required by the system used.
If using an RS422/LVDS type flat panel sensor, then the National Instruments Measurement & Automation Explorer (MAX) and NI IMAQ PCI-1424 grabber card can be used to evaluate the basic functions.
No software is supplied with flat panel sensors. Customers must create software according to the application. However, the basic functions of flat panel sensors can be evaluated by using the "Hipic U8913-01" software that is available for use with Hamamatsu digital cameras. Please contact our sales office when you are interested in the Hipic.
Specifications differ depending on the types, so see the datasheets for details.
Setting the X-ray dosage to a low level within the detectable range or turning off X-rays so that the flat panel sensor is not exposed to radiation except during image capture will extend the flat panel sensor service life. Another effective method is using a pulsed X-ray source. Also recommended is periodically acquiring dark images and shading images for image correction according to fluctuations in the ambient temperature and sensor temperature.
The following criteria are used to determine the life end in our lifetime test.
*Average value of dark increase is 500 GL or less in internal mode. (12-bit output products)
*Sensitivity is 50% or less of the initial value.
*Number of defect lines is double or more the spec value.
We have a demonstration room available for customer use. We will also acquire X-ray images of samples sent to us from customers. Demo devices are also available for loan to customers. Please contact our sales office.
It looks like you're in the . If this is not your location, please select the correct region and country below.
You're headed to Hamamatsu Photonics website for GB (English). If you want to view an other country's site, the optimized information will be provided by selecting options below.
For modern websites to work according to visitor’s expectations, they need to collect certain basic information about visitors. To do this, a site will create small text files which are placed on visitor’s devices (computer or mobile) - these files are known as cookies when you access a website. Cookies are used in order to make websites function and work efficiently. Cookies are uniquely assigned to each visitor and can only be read by a web server in the domain that issued the cookie to the visitor. Cookies cannot be used to run programs or deliver viruses to a visitor’s device.
Cookies do various jobs which make the visitor’s experience of the internet much smoother and more interactive. For instance, cookies are used to remember the visitor’s preferences on sites they visit often, to remember language preference and to help navigate between pages more efficiently. Much, though not all, of the data collected is anonymous, though some of it is designed to detect browsing patterns and approximate geographical location to improve the visitor experience.
Certain type of cookies may require the data subject’s consent before storing them on the computer.
This website uses two types of cookies:
There are two ways to manage cookie preferences.
If you wish to restrict or block web browser cookies which are set on your device then you can do this through your browser settings; the Help function within your browser should tell you how. Alternatively, you may wish to visit www.aboutcookies.org, which contains comprehensive information on how to do this on a wide variety of desktop browsers.
Occasionally, we may use internet tags (also known as action tags, single-pixel GIFs, clear GIFs, invisible GIFs and 1-by-1 GIFs) at this site and may deploy these tags/cookies through a third-party advertising partner or a web analytical service partner which may be located and store the respective information (including your IP-address) in a foreign country. These tags/cookies are placed on both online advertisements that bring users to this site and on different pages of this site. We use this technology to measure the visitors' responses to our sites and the effectiveness of our advertising campaigns (including how many times a page is opened and which information is consulted) as well as to evaluate your use of this website. The third-party partner or the web analytical service partner may be able to collect data about visitors to our and other sites because of these internet tags/cookies, may compose reports regarding the website’s activity for us and may provide further services which are related to the use of the website and the internet. They may provide such information to other parties if there is a legal requirement that they do so, or if they hire the other parties to process information on their behalf.
If you would like more information about web tags and cookies associated with on-line advertising or to opt-out of third-party collection of this information, please visit the Network Advertising Initiative website http://www.networkadvertising.org.
We use third-party cookies (such as Google Analytics) to track visitors on our website, to get reports about how visitors use the website and to inform, optimize and serve ads based on someone's past visits to our website.
You may opt-out of Google Analytics cookies by the websites provided by Google:
We inform you that in such case you will not be able to wholly use all functions of our website.